# How to measure with Screw Gauge.

Aim: To measure the i)Thickness of the Glass Plate ii) Diameter of the metal wire iii) Volume of the given Glass Plate.

Apparatus : Screw Gauge , Glass Plate and Metal wire .

Description : Screw Gauge consists of U shaped metallic frame.To one side of this U frame a long hallow cylindrical tube with a nut inside it, the inner side of cylindrical nut contains a uniform thread cut in it.On the other side of U frame a fixed stud S_1 with a plane face is attached.

A screw S_2 is fitted in the cylindrical nut.One side of the screw S_2 has a plane face similar to that of stud S_1. The faces of S_1 and S_2 are plane and parallel to one another. The other end of the screw S_2 carries a milled head ‘H’ attached to a cap ‘C’ with a sloping edge. When the head H is rotated, the screw moves ”to and fro” in the nut.The milled head H is provided with a safety device ‘D’ to rotate the head H.When the object is held between the stud S_1 and screw S_2 and the head H is rotated using the safety device (D), it produces crackling sound when optimum pressure is applied on the object.

The outer surface of long cylindrical nut consists of a thick horizontal line ‘P’ parallel to the axis of cylindrical tube.This line ‘P’ is called Index line. Along the index line a scale is graduated in millimeters.This scale is called Pitch Scale.On the sloping edge of the cap ‘C’ a circular scale is graduated, which consists of 100 equal divisions, this scale is called Head scale.

Theory : The screw gauge works on the principle of screw.

When we rotate the head ‘H’ by means of safety device ‘D’ through one complete rotation, the distance moved by the screw for every complete rotation is constant. This constant distance moved by the screw for one complete rotation of head ‘ H ‘ is called Pitch of the screw.If the head scale has 100 equal divisions, then the distance moved by the screw for even 1/100 of a complete rotation can be measured accurately,this is called Least count of screw gauge.

Therefore Least count (L.C) = \frac{Pitch of the screw}{Number of divisions on Head scale} .

Procedure :First we have to determine the least count of the given Screw gauge.

To determine the least count of the screw gauge, the head ‘H’ is rotated through certain (say 5) number of complete rotations.The distance moved by the sloped edge over the pitch scale is measured.

Now substitute these values in the formula of pitch of the screw = \frac{Distance moved by sloped edge over the pitch scale}{Number of rotations of the screw}.

Least count L.C = \frac{Pitch of the screw}{Number of divisions on Head scale} .

Now check whether the given screw gauge has any ZERO ERROR or not. To determine the ZERO ERROR, the head H is rotated until the flat end of the screw S-2 touches the plane surface of the stud S_1 (do not apply excess pressure) i.e we have to rotate the head only by means of safety device ‘D’ only.

zero-error

When S_1 and S_2 are in contact,the zero of the head scale perfectly coincides with the index line as in Fig-(a). In such case there will be no ZERO ERROR and no correction is required.

When S_1 and S_2 are in contact,the zero of the head scale is below the index line as in Fig(b), such ZERO ERROR is called positive ZERO ERROR, and the correction is negative.

When S_1 and S_2 are in contact,the zero of the head scale is above the index line as in Fig(c) , such ZERO ERROR is called negative ZERO ERROR, and the correction is positive.
When S_1 and S_2 are in contact,the zero of the head scale is above the index line as in Fig(c) , such ZERO ERROR is called negative ZERO ERROR, and the correction is positive.

a) Determine the thickness of glass plate : The given object glass plate is held between the two parallel surfaces of fix stud S_1 and screw tip S_2. Note the completed number of divisions on pitch scale, which is called PITCH SCALE READING (P.S.R). The number of the head scale division coinciding with the index line is noted, which is called OBSERVED HEAD SCALE READING n’. If the given screw gauge has ZERO ERROR (x) the correction is made by adding or subtracting the ZERO ERROR (x) from the OBSERVED HEAD SCALE READING n’.The corrected value (n’-x) or (n’+x) is called the HEAD SCALE READING (H.S.R) n.

To calculate the fraction the H.S.R (n) is multiplied by the least count (L.C).

Thickness of the Glass plate = Total reading = P.S.R +n\times L.C - - - - - (1)

Changing the position of glass plate , 5 readings should be taken, and recorded in the table-1. Every time calculate the total thickness of the glass plate using equation (1).

Average of the 5 readings of the glass plate should be calculated, to get the average thickness(t) of the given glass plate.

To calculate the fraction the H.S.R (n) is multiplied by the least count (L.C).

Diameter of the given wire = Total reading = P.S.R +n\times L.C

Changing the position of metal wire, 5 readings should be taken, and recorded in the table-2. Every time calculate the total diameter (d) of the metal wire using equation (1).

Average of the 5 diameter of the metal wire should be calculated, to get the average diameter(d) of the given metal wire.

Radius (r) of the metal wire =\frac{d}{2} mm.

Precautions : i ) Pitch scale reading (P.S.R) should be taken carefully without parallax error ii ) Head scale reading (H.S.R) should be taken carefully without parallax error iii )Screw must be rotated by holding the safety device ‘D’ iv ) Do not apply excess pressure on the object held between the surfaces S_1 and S_2.

v ) The screw is rotated in only one direction either clock wise or anti-clock wise to avoid the back lash error.

Observations : i ) Zero error =

ii) Zero correction = mm

iii ) Distance moved by the head for 5 complete revolutions = mm

iv ) Number of head scale divisions =

v) Pitch of the screw = \frac{Distance moved by sloped edge over the pitch scale}{Number of rotations of the screw}.

vi) Least count (L.C) =\frac{Pitch of the screw}{Number of divisions on Head scale} .

c ) Volume of Glass plate (v) : The length ( l ) , breadth ( b) are determined using vernier calipers and thickness ( t ) of the glass plate is determined using screw gauge. The values of l ,b and t are substituted in the equation of volume V = ( l )( b )( t ) mm^3

Reference: http://gyaunnrraje.wordpress.com/2009/02/18/screw-gauge/

## Related Articles

#### How to prevent acne scars?

This article explains various measures which keep acne under control. This decreases inflammation and consequently chances of formation of acne scars too decreases. Actions like not picking at skin, proper hydration, using sunscreen lotions, proper skin care regime etc prevent formation of acne scars.

#### How To Control Anger?

Have you ever been angry and later on felt sad for being angry on some one or something? Do you want to control your anger ? Is anger good or not ? The following article will discuss about anger and measures through which we can control our anger.

#### How to remove blood stains from delicate fabrics and specific surfaces?

This article explains various tips and methods to get rid of blood stains from delicate fabrics like silk, satin, woolen, linen etc and from specific surfaces like concrete, hardwood floors, mattress, quilt, leather, carpets etc. These methods used can make them free from blood stains.

#### How to become a pathologist with top hospitals in India

Are you looking for a career in medicine as a clinical pathologist? Need to know how to start a pathology lab? This article provides complete information on the different specializations in pathology and a career guide on starting a career as a pathologist. You will also get a general idea on the approximate costs of setting up your own path lab.

More articles: How to